DAVID Live is a revolutionary new app geared to optimize clinician/client experience. The DAVID Live app in combination with the Spectrum™ eyesets is perfect for a clinician practicing neurofeedback, talk therapy, hypnosis, or brain stimulation where the therapist requires control. The clinician has full control over dual-frequencies, intensities, and color. The client also has self-guided light intensity control using two buttons located on the Spectrum™ eyeset screen, which can also be disabled by the therapist.
2. Double click on the DAVID Live eyeset icon to begin.
To control multiple eyesets at once, hold Shift and select desired eyesets and press the Enter key.
Right-click on a device and select Rename to assign a custom name.
Due to manufacturing variances, the LEDs in some eyesets may not produce a uniform white from the factory. Eyeset calibration allows the color of each LED to be adjusted to correct for these variances.
Right-click on an eyeset and select Calibrate to start eyeset calibration.
The DAVID Live app will offer to automatically install any firmware updates when a product is selected. This process takes approximately 30 seconds and is highly recommended.
Dual mode operates the left and right channels at independent frequencies.
Sync mode operates the left and right channels at the same frequency and uses phase control to ensure both channels are in phase.
Alternate mode operates the left and right channels at the same frequency and uses phase control to ensure both channels are 180° out of phase.
When Randomization is enabled, the device generates random ±1 Hz frequency offsets for the left and right channels at 2 to 5 second intervals.
Live Mode applies changes to the stimulation settings automatically. The transition setting can be used to add a short fade to every change for smoother adjustments.
Manual Mode applies changes to the stimulation settings on demand. The transition setting can be used to add a short fade to every change for smoother adjustments.
The Timer feature freezes the current stimulation settings and performs an automatic soft-off when the countdown expires.
Pause and Resume
The Pause button freezes the current stimulation settings and performs a 10 second fade to black. While paused, the stimulation settings can be changed without affecting the device. Resuming from pause will perform a 10 second soft-on using the current stimulation settings.
Visual Stimulation Settings
Classic generates synchronized semi-square waves using the selected colors and channel frequencies.
Sine generates synchronized sine waves using the selected colors and channel frequencies.
Triangle generates synchronized triangle waves using the selected colors and channel frequencies.
Remote Level Control
The Brightness button allow the stimulation levels on the device to be remotely viewed and adjusted by the operator.
The User Adjustment Lock button can be used to disable the volume and brightness controls on the device.
We always recommend having a glass of water 20 minutes before every entrainment session.
Make yourself comfortable and try to limit interruptions.